Instructions for Use
About MedLite ID Smart-Lites
The MedLite goal is to help reduce infusion Adverse Drug Events, speed access to the medication port, and reduce infectious disease exposure time; all while providing a smart, simple, safe solution for the nurse and the patient that creates standardization within the hospital.
Location of MedLite ID Smart Lites may communicate hospital-specific infusion scenarios. Individual hospital infusion safety goals and training can determine the best use cases to assist in preventing infusion ADE. Common use cases selected by hospitals include identification of the KVO line or high-risk drug lines such as vasopressors or heparin. As the patient journey moves them from department to department, this simple safety standard protects against infusion ADE throughout the hospital, for every shift, and in all lighting conditions.
MedLite is recommended for patients with two or more IV lines. The Lites are single-use, disposable devices and will last about 96 hours, tied to INS line-change guidelines. Lites may be disposed of in non-hazardous waste.
Example Use Case: MedLite ID Identification of the KVO Line
How to Activate Smart-Lites
Each 3-Lite set is packaged within a detailed IFU and includes a label with a URL link and QR code to this training video. The label should be placed on the IV line just under the first MedLite Smart-Lite.
STEP 1: To ACTIVATE the Lites, remove each Lite from the strip, leaving each Lite’s clamshell open. Batteries will activate and start to blink during auto-pairing, then turn off.
STEP 2: Before connecting the SSL to your patient, ATTACH the Lites to the line, pressing each into the center grooves at the top, middle, and bottom of each LITE, ensuring no kinks. Do not connect the Lites to an already hanging IV line.
STEP 3: To LOCK the Lites, close the top of each light until it clicks.
How to Apply Smart-Lites, One Example:
Apply the first Lite below the drip chamber at eye level.
Apply the second Lite below the infusion pump (or the middle of the IV line if applying to a gravity set). On lines 3.0-4.2mm outer diameter, the Lites will be “firm fit”; on lines less the 3.0, they will be “slip fit.”
Apply the third Lite prior to the venous access catheter port.
Hang this IV line per standard hospital protocol.
To use the Lites, press the button on any one Lite to illuminate all three for about 50 seconds.
Lites turn off automatically and last about 96 hours. Tied to INS line-change guidelines.
Lites are non-sterile and may be disinfected with 70% IPA wipe, Sani-WipesTM used as directed, or germicidal bleach
Lites may be disposed of in non-hazardous waste.